- Dissertation Council
- Products & Technologies
- IWTMC IV
This is the archive site of the NIIC SB RAS.
The information on this site is not up to date.
The current website of the NIIC SB RAS is available here:
Generators of standard concentrations (GSC) are intended for preparation, dosage and inlet of vapour-gas mixtures of organic and inorganic compounds into a gas analyzer.
Organic and inorganic compounds intercalated into fluorinated graphite (CIFG) are used as a source of vapour in GSC. Fluorinated graphite is inert toward a wide range of substances; playing the part of a chemical container for reactive substances, it leads to a decrease in their saturated vapour pressure.
After metrological certification, the GSM can be used as a secondary means for certifying and calibration of gas analytical devices. GSC can be used under field conditions..
|Amount of CIFG per one device [g]||0,2—1|
|Capacity of CIFG (content of substance to be analyzed [mas. %])||20|
|Maximal error of gas-vapour mixture preparation [%]||5|
|Time of achieving the working mode under temperature changes [hour]||
|Time of achieving the working mode after previous analysis [min]||1—60|
|Time of stable operation [year]||1|
Universal character of the method of obtaining vapour-gas mixtures for a wide range of easily volatile and gaseous substances.
Ecology, industry, medicine, modern technologies, instrument-making industry.
Prototypes of devices (GSC). At present, GSC are used under laboratory conditions:
Patents granted in Russian Federation and Europe.
The invention was awarded Gold Medal at the Exhibition in Brussels in 1995.
Nikolaev Institute of Inorganic Chemistry SB RAS
Acad. Lavrentyev ave., 3, Novosibirsk, Russia, 630090
Scientific secretary of Institute
Gerasko Olga Anatolievna, Doctor of chemical sciences
Phone: +7(383) 330-94-86